Frozen Fat Dissolving Apparatus
Too much fat tissue in the body can damage a person's appearance and exercise ability, and may also cause various diseases. Excess fat tissue can be located in various parts of the body, such as: thighs, buttocks, abdomen, knees, back, face, arms and other areas. And too much adipose tissue will form an unsightly appearance like a honeycomb. Excess body fat may be related to health risks, so it is necessary to effectively control or eliminate excess fat.
To reduce the body’s stubborn fat, in addition to using RF heat, laser-assisted liposuction and high-intensity focused ultrasound to accelerate its metabolism, it can also use a new cryo-fat technology to thin the fat layer to achieve self-cultivation and shaping effects .
The latest freeze-melting method is the opposite of heating method, which is a new method that is more effective than other non-invasive methods. A revolutionary new concept of fat dissolving, put into use has proved to be a major breakthrough in the battle of fat dissolving.
For those who have a reasonable diet and exercise regularly but still can’t get rid of local fat, the latest cryo-fat technology is undoubtedly a godsend. Whether it is for areas with denser fat or smaller areas of fat, such as loose fat on both sides of the waist above the buttocks, belly and back fat, this freezing fat melting technology can produce stunning results.
A) The treatment device belongs to class I equipment according to the anti-shock type;
B) The power supply for the treatment device is AC220V±22V; 50Hz±1Hz;
C) The treatment device has a BF type application part;
D) The therapeutic device does not have a signal input part and a signal output part;
E) The treatment device is an ordinary device;
F) The operation mode of the therapy device is continuous operation mode.
Scope of application
Slimming: Slimming the face, neck, limbs, buttocks and abdomen, firming up the face, chest, and buttocks, and recovering body shape after delivery and liposuction.
Pregnancy lines: Effectively remove stretch marks.
Slimming: Reduce cellulite, reduce cellulite, and improve sagging skin.
Skin tightening: Strengthen and tighten skin, increase skin elasticity, improve cellulite, postpartum stretch marks, etc.
1.4.1 Environmental conditions for use
A) Ambient temperature: 5～35℃;
B) Relative humidity: ≤80%;
C) Atmospheric pressure: 80KPa～106KPa.
1.4.2 Environmental requirements for packaging, transportation and storage
A) Ambient temperature: 0～40℃;
B) Relative humidity: ≤90%;
C) Atmospheric pressure: 80KPa～106KPa;
D) The transportation of the treatment device should be carried out after packaging, and pay attention to "fragile", "upward", and "fear of rain";
E) The treatment device should be stored in a room with no corrosive gas and well ventilated.
System composition and performance parameters
Remarks: When the temperature of the treatment head does not reach 6°C or more during the heating process, the system suspends the heating process and waits to be directly added to the refrigeration process; when the temperature difference between the two sides of the refrigeration chamber reaches 3°C during the refrigeration process, the system automatically Stop this work, wait for the system to recover for about one minute, and then try to treat again.
Using the feature that the triglycerides in fat will be converted into solids at a specific low temperature, a precisely controlled low-temperature wave is transmitted to the designated fat-dissolving site through a non-invasive transmitter to target and eliminate fat cells. After the fat cells are exposed to cold, they will start a natural process of decomposition and removal, causing the fat layer to gradually become thinner. Thus, through the normal metabolic process, the fat layer is gradually reduced to achieve the purpose of local fat dissolution.
Main performance parameters
1. Power supply: 220V/50Hz or 110V/50Hz
2. Display: 8-inch color touch screen
3. The output temperature of the refrigeration device: low temperature -5～5℃; high temperature 37～39℃
4. The output pressure of the refrigeration device: negative pressure 10～80Kpa
5. Instrument cooling liquid: pure water
6. Timing time: 1～60 minutes
1 Insert the key into the key switch, turn it 90° clockwise, turn on the power of the instrument, and enter the main operation interface
① Real-time display of the status of the treatment head
T: Display temperature. P: Shows the pressure at the treatment head.
② Display the type of treatment head connected to the machine
S: Trumpet. M: Medium. L: Large size. (Temporarily only provide L-shaped treatment head)
③ Set the heating temperature of the treatment head
④ Set the cooling temperature of the treatment head
⑤ Set negative pressure value
⑥ Set timed treatment time
⑦ Remaining time of treatment
⑧ Start/Stop button (This instrument is a dual-head therapy, two therapy heads can be used at the same time, and the left and right setting interface functions are the same.)
2 treatment head use
① Increase the negative pressure value.
② Reduce the timing time.
③ Reduce the negative pressure value.
④ Increase the timing time.
⑤ "Enter" confirmation key. After adjusting the parameters of the treatment head, press this button, the parameter value is successfully modified,
Incoming instrument operation interface is displayed synchronously. (Note: If the parameter value is adjusted and the confirmation key is not pressed, the parameter still uses the value before modification.)
Parameter display window. The first two digits represent negative pressure, and the last two digits represent timing time.
3 Start treatment
After confirming the settings of each parameter, click the "Start" button on the interface or press the start button of the treatment head handle (see the figure below), and the treatment head starts to work. The working program is set to start heating up to the set high temperature after a few seconds, and then the cooling reaches the set low temperature.
4 When the scheduled treatment time is reached, the computer automatically turns off the treatment light and the treatment is completed.
5 After the treatment time is up or the treatment is stopped, the treatment head display window will flash "00:00". You need to press the treatment head handle button once to release the air pressure before the next treatment starts.
6 Turn off the key switch and cut off the power supply of the host.
7 Clean the external treatment cover of the treatment head
After the treatment is completed, the treatment mask that is in contact with human skin must be cleaned and disinfected.
Precautions and contraindications
1 Treatment head installation
Before inserting the treatment head, be sure to press down the bayonet shrapnel on the socket (see the picture below) to ensure that the middle button is raised before inserting the treatment head.
Then align the interface end of the treatment head with the treatment head socket on the machine and insert it. When you hear the sound of the shrapnel popping up, it means that it has been inserted.
When removing the treatment head, press and hold the round button on the treatment head interface (that is, release the internal bayonet shrapnel) and pull it out.
2 Fill the coolant
The coolant in the internal coolant tank of the machine has been discharged during transportation (remember to cover the air holes with air hole caps during transportation), and liquid injection must be carried out before use.
Unscrew the air hole before injecting the coolant. Align the injection tube with the injection hole on the back and inject about 2000ML of pure water before starting to use the instrument.
When the machine starts treatment, the vent should also be opened, and the vent should be screwed on only when the coolant is prevented from leaking during transportation.
3 rows of coolant
The drain hole is at the bottom of the machine. Unscrew the nut at the bottom to start draining the coolant.
Before using this machine, please read this chapter carefully to understand and master the safety precautions described in this chapter.
Adjusting or using the machine not in accordance with the requirements of this chapter may cause personal injury or damage to the machine. Therefore, medical personnel and technicians who operate or maintain the machine must be familiar with the safety requirements of the machine before operating the machine.
2 Personnel safety protection
2.1 In the design process of the treatment device, the safety of medical staff and patients has been considered to the utmost extent, and the following safety precautions have been taken:
A) If necessary, the power supply can be quickly cut off by the emergency stop switch to turn off the host;
B) The key switch can prevent non-professional operators from misusing the machine.
2.2 Light radiation may damage the eyes or skin (especially eyes). All necessary protective measures should be taken during use to avoid direct or scattered exposure to the eyes or skin.
2.3 The safety of patients mainly depends on well-trained physicians and appropriate treatment rooms. Before treatment, the doctor should explain the treatment process and precautions to the patient in detail, so that they can fully understand the treatment principle and actively cooperate with the treatment. At the same time, when medical staff perform treatments on patients, they must also do their own safety protection, mainly eye protection, and should not look directly at the output light during the entire treatment operation.
2.4 The walls, roof, and ground of the treatment room should be free of light reflectors, such as mercury mirrors, large-area high-gloss metal objects, etc.
2.5 During the treatment, only necessary and well-trained personnel should be allowed to approach the machine.
3 Environmental conditions for use
The use of flammable anesthetics or oxidizing gases (such as nitrous oxide and oxygen, etc.) in the treatment room must be avoided.
Some materials, such as cotton wool, will be ignited by the high temperature generated by the equipment in normal use when it is rich in oxygen.
Solvents and flammable solutions used for cleaning and disinfection should be volatilized before using this equipment.
Attention should be paid to the danger of ignition of internal gas.
Product maintenance and troubleshooting
Method of exclusion
Can not boot
1. Poor contact of the power cord interface;
2. The air switch is in the "OFF" state.
1. Reconnect the power cord.
2. Push the air switch to the "ON" state.
The screen keeps prompting "water lacking" for a long time after booting.
1. There is no water in the water tank;
2. There is very little water in the water tank.
1. Add enough water as required;
2. Add enough water as required.
During the treatment process, the treatment time is often not reached, the treatment is automatically terminated, and the LCD temperature display color changes from black to red.
The performance of the instrument's refrigeration system is degraded.
Contact the manufacturer for update and maintenance.
Packing list and packaging
1.The host computer, one unit
2.Stents stents stents
3.The treatment head According to the provisions of the order contract
4.Power cord One piece
6.Qualification certificate One copy
The instrument is wrapped in a protective film and foam and placed in a carton with foam board for shock isolation.